F2 Set Number of
Points 
Enter the number of points. The number of measurement point
directions and number of points determine the total number of locations to
collect data. For example, if the measurement direction is XYZ and 1,024 points
are defined, a total of 3,072 locations (1,024 X, 1,024 Y and 1,024 Z) are
available in the job. The default is 20. 
F3 Set Starting
Point 
Enter the first measurement point to display when you
collect data. If a job has 20 measurement points, the starting point could be
from 1 through 9,979. If the starting point is 9,979, the second point number
would be 9,980. The default is 1. 
F4 Set Coordinate
Sys 
Use the up and down arrow keys to select
Rectangular (X, Y, Z),
Cylindrical (Radial, Theta, Axial), or
Spherical (Radial, Theta, Phi). The
default is
(X, Y, Z).
Note: ME'scope VES™ uses X, Y, Z.

F5 Set Meas
Direction 
Use the up and down arrow keys to select the direction. The
available directions depend on the coordinate system defined for the job in
F2 Set Coordinate Sys. The default is XYZ.
For example, if you selected rectangular (XYZ), the possible measurement
directions are X, Y, Z, XY, XZ, YZ, and XYZ.

F10 Set Meas Input 
Set the input for the displayed measurement direction
(orientation). This option does not appear if you have only two inputs. If you
disable the input for this direction, "???" displays.

F11 Set Meas Input 
Set the input for the displayed measurement direction
(orientation). This option does not appear if you have only two inputs. If you
disable the input for this direction, "???" displays.

F12 Set Meas Input 
Set the input for the displayed measurement direction
(orientation). This option does not appear if you have only two inputs. If you
disable the input for this direction, "???" displays.
