By default, the analyzer is setup for CSI 404 tachometers.
Pseudo Tach must be disabled for the Balance program. When the pseudo tach is disabled, a standard 1x tach is produced.
Balance main menu, press
The Tachometer Setup screen appears.
Set the following options as necessary.
Option Description F2 Pseudo Tach Disable for the Balance program. When the Pseudo Tach is disabled, a standard 1x tach is produced. F5 Tach Power Enable the analyzer to power the tachometer. The default is to provide power to the tachometer. F7 Set Trigger Edge
Set the trigger to occur on the rising edge or falling edge of a waveform. The default is Falling Edge.
F8 Set Trigger Level Enter the value between -100 and 100 required to start a tachometer pulse. For Level measurement, set the input that is the source of the trigger. The default is 2.0 Volts. F9 Set Edge Delay Enter the number of seconds between 0 and 16 that occurs between each tachometer pulse. This helps prevent double triggering. The default is 0.0 seconds. F10 Show RPM F10 Hide RPM View or hide the last RPM the tachometer read and the time between each reading. A last time value of 0 indicates the tachometer is continuously reading the RPM. F12 Set Defaults Restore the default values for all tachometer setup options on the screen.
- Press F6 Save / Recall Setup to save the setup, or press Enter to use the setup without saving it.