- On the analyzer, press and ensure the default print mode is set to Send to PC.
- Ensure the same connection type is set on the analyzer and in Data Transfer. The Ethernet and Wireless options may require additional setup to connect to the desired computer. See the Ethernet and Wireless sections and your IT group for more information.
- Use the appropriate cable to connect your analyzer to a computer where AMS Machinery Manager is installed, if you are using a wired connection.
- On the computer, open Data Transfer.
Display your plot in ODS/Modal.
You can press F7 Display Data on the ODS/Modal main menu, or collect data and view the plot.
F7 Print Plot.
The plot appears in AMS Machinery Manager. You can print the plot as an image file on the computer.